Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||13 June 2010|
|PDF File Size:||7.10 Mb|
|ePub File Size:||15.15 Mb|
|Price:||Free* [*Free Regsitration Required]|
4191A RF IMPEDANCE ANL
I have worked with the A and, as I recall, the commands are nothing like any other instrument Hp4191q have seen. Most Active Software Boards: The forums emoticons took over for a bit there.
Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. Message 6 of 6.
Agilent / HP 4191A RF Impedance Analyzer
Message 2 of 6. However, rather than being returned the data at MHz, I am returned the data at whatever frequency hp4119a analyzer is currently set at. Should I be looking at doing it with LabView or something else? Message 5 of 6.
Message 3 of 6. Would you mind posting the code you used on the A?
I have not worked with that instrument, but the Agilent web site has the operating and service manual ho4191a. I am able to communicate with the device through the GPIB interface, and am also able to retrieve information. If not, what would creating a driver for it entail?
It’s 84MB, so I hope you have a fast link. Message 1 of 6.
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
Good luck with it In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:. In the code above, MHz is an arbitrary frequency.
Guess you never know until you try! Matt – you’re right, the commands for the A are completely different from the – probably because of its age. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: IIRC, the sweep parameters were something like this: When I gave it a try, though, it worked fine. Message 4 of 6. Accepted by topic author Matt C. I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so hp4191aa anything I’ve ever seen before that I assumed it was referring to something different.
I suspect that this difference might be causing my manyal.
Solved: GPIB Coding for the HPA Impedence Analyzer – Discussion Forums – National Instruments
I think the would probably take the old commands, but the ALSO has new commands. You really need the manual. Thanks for your help guys. Has anyone done anything h4191a interfacing with the A since Matt C’s last post?
Sorry, that last line of code there is supposed to read: