BRAND offers the ideal pipette for every hand: BRAND’s new Transferpette® S. Transferpette® S models are are the perfect manual pipettes for demand-. Transferpette® electronic. F I R S T C L A S S · B R A N D. Gebrauchsanleitung. Operating Manual. Mode d’emploi. Instrucciones de manejo. 3. -8/ . Testing Instructions (SOP) testing of the piston operated pipette Transferpette®. BRAND additionally provides a calibration service (see chapter 7).
|Published (Last):||4 February 2008|
|PDF File Size:||11.79 Mb|
|ePub File Size:||5.96 Mb|
|Price:||Free* [*Free Regsitration Required]|
It is the responsibility of the user of this pipette to consult and establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Page 9 Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. Ensure that the tip is securely seated. By pressing ‘print’ button you will print only current transfepette. Discharge sample a Place the pipette tip against the wall of the vessel.
Inspect the piston and O-Ring-seal for contamination. Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Page 17 Inspect the pipette tip cone for damage.
Pipetting Pipetting 5 ml and 10 ml instruments should only be used with the PE filter installed see page Inspect the pipette tip cone for damage. Pipetting Ejecting the tip Hold the pipette shaft over a suitable disposal container and press the tip ejection key to the stop. The form can also be downloaded from www. In order for the liquid to reach its end position, leave the tip immersed for a few seconds.
Ask your supplier or manufac- turer for the form. For further information, refer to the accuracy table on pages 37 and Don’t have an account? Laboratory Equipment Brand dispensette Operating Transferpettr pages. Limitations of Use Purpose The pipette is an air-displacement system for pipetting aqueous solutions with medium density and low to medium viscosity. Eject the pipette tip.
Enter text from picture: Also See for Transferpette S Operating manual – pages.
Just send in the instruments to be calibrated, accompanied by an indication of which kind of calibration you wish. If the pipette operation is clearly inaccurate, or if the instrument must be adjusted for solutions of different densities and viscosities or specially- shaped pipette tips, adjustments can be made using the Easy Majual Technique.
Warranty We shall not be liable for the consequences of improper handling, use, servicing, operating or unauthorized repairs of the instrument or the consequences of normal brqnd and tear especially of wearing parts such as pistons, seals, valves and the breakage of glass as well as the failure to tarnsferpette the instructions of the operating manual.
Safety Instructions Please read the following carefully! Adjustment — Easy Calibration The instrument is permanently adjusted for aqueous solutions. Pipette shaft Tip cone Fig.
BRAND TRANSFERPETTE S OPERATING MANUAL Pdf Download.
Parts will differ slightly depending on nominal volume of instrument. Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months. Don’t lay the instrument horizontal when the tip is filled. Consult the manufacturer for use in temperatures outside of this range.
Comments to this Manuals Your Name. Got it, continue to print. Inspect the piston and seal for contamination. To print the manual completely, please, download it. Hold the instrument vertically, with the sample in the tip for approximately 10 s.
Pipette tips are disposables items! Trahsferpette of Contents Add to my manuals Add. Autoclave the complete pipette without any further disassembling. Safety Instructions Safety Instructions Please read the following carefully!
BRAND Transferpette S Operating Manual
Mznual the tip Use the correct tip according to the volume range or the color code. ISO prescribes rinsing the pipette tip once with the sample liquid prior to the actual pipetting process. This instrument may sometimes be used with hazardous materials, opera- tions, and equipment.
Checking The Volume Checking the Volume Depending on use, we recommend inspection of the instrument every 3 to 12 months.
Set nominal volume Set volume to the maximum volume indicated on the instrument see page 32 for procedure.